ATTENTION TO THOSE WITH AN ADJUSTABLE SHUNT!! IMPORTANT iPad2 and MAGNETIC SHUNTS DO NOT MIX!
Dear Members: Today we received the official information from Medtronic Neurosurgery on the use of the Apple iPad 2 and persons with programmable shunts.
The email is from Jeffrey Henders, VP of Quality and Regulatory Affairs at Medtronic.
Below is a portion of the email.
"Dear Mr. Illions, On December 9, 2011 we received a question from you regarding the use of an Apple iPad 2 for individuals with an implanted Strata Valve. A sample iPad 2 was evaluated for magnetic strength. The left front edge of the iPad 2 was found to have a magnetic strength of 306 Gauss and the rear speaker measured 456 Gauss. The Strata family of valves is designed to be adjusted with a permanent magnet of at least 90 Gauss, directly against the valve. This indicates that the iPad 2 has the potential to affect the Strata Valve setting if the areas having the above measured fields are placed directly adjacent to the implant location of the Strata Valve. Inadvertent adjustments by external magnets are a known complication with adjustable valves. Therefore, the same precautions that are applied to all magnets should also be applied to the iPad 2"
I saw your posting and decided to contact Medtronic myself to see if I could get some clarification on how careful you would need to be with an iPad and a programmable shunt. I talked to Leanne Lintula a Senior Product Manager at Medtronic Neurosurgery and she was kind enough to send me an email overview of our conversation. Below is the message in its entirety (when she refers to the "information noted below" she is referring to when to I quoted this forum posting.
Per our conversation this morning, I am writing to confirm that we did discuss the magnetic strength of the IPad 2. As I mentioned, the information noted below was a preliminary notice we did send out to the field. Since that time, we have quantified the safety distance needed for the iPad 2. That is, as long as the iPad 2 is kept a minimum of 2" from the valve it will not interfere with the valve setting.
We are currently updating our patient information. This information is forthcoming and you will be able to download the file from the Medtronic website at www.medtronic.com.
As we discussed, it will be a couple of weeks before the final copy is ready, so please look for it then.
Senior Product Manager, Hydrocephalus
125 Cremona Drive
Goleta, CA 93117 USA
The Content on this Site is presented in a summary fashion, and is intended to be used for educational and entertainment purposes only. It is not intended to be and should not be interpreted as medical advice or a diagnosis of any health or fitness problem, condition or disease; or a recommendation for a specific test, doctor, care provider, procedure, treatment plan, product, or course of action. Med Help International, Inc. is not a medical or healthcare provider and your use of this Site does not create a doctor / patient relationship. We disclaim all responsibility for the professional qualifications and licensing of, and services provided by, any physician or other health providers posting on or otherwise referred to on this Site and/or any Third Party Site. Never disregard the medical advice of your physician or health professional, or delay in seeking such advice, because of something you read on this Site. We offer this Site AS IS and without any warranties. By using this Site you agree to the following Terms and Conditions. If you think you may have a medical emergency, call your physician or 911 immediately.