Determination of reliability criteria for liver stiffness (fibroscan)
Determination of reliability criteria for liver stiffness evaluation in a cohort of 1165 patients
Background and aims: The manufacturer of Fibroscan has defined reliability criteria for liver stiffness evaluation (LSE): ≥10 valid shots with ≥60% success rate and IQR/median ratio (IQR/M) ≤30%. In the present study, we aimed at providing a first-ever assessment of the validity of these LSE reliability criteria. Methods: 1165 patients with chronic hepatitis, liver biopsy, and LSE by Fibroscan were included. Fibrosis classification (F0/1, F2/3, F4) was defined from LSE results according to diagnostic cut-offs published by Friedrich-Rust (Gastroenterology 2008). Results: Causes of chronic hepatitis were: viral hepatitis (76.3%), alcohol (12.6%), NAFLD (2.8%), and other (8.3%). 75.3% of LSEs were reliable according to the manufacturer’s criteria. AUROCs of significant fibrosis (F≥2), severe fibrosis (F≥3), and cirrhosis (F4) were not significantly different between reliable and unreliable LSEs, respectively: 0.836±0.014 vs 0.789±0.027 (p=0.122), 0.882±0.011 vs 0.857±0.023 (p=0.327), and 0.915±0.012 vs 0.906±0.021 (p=0.710). Significant fibrosis was independently associated with LSE result (1st step, p <10-3), age (2nd step, p <10-3), and IQR/M (3rd step, p=0.028). Severe fibrosis was independently associated with LSE result (1st step, p<10-3), IQR/M (2nd step, p=0.007), and age (3rd step, p=0.007). Cirrhosis was independently associated with LSE result (1st step, p <10-3), biopsy length (2nd step, p=10-3), and IQR/M (3rd step, p=0.011). AUROCs for significant fibrosis, severe fibrosis, and cirrhosis were significantly higher when IQR/M was ≤0.10. In LSE 0.30. In LSE >13.0kPa, the diagnostic accuracy significantly decreased when IQR/M was >0.10. 17.3% of patients had an IQR/M ≤0.10, 68.1% had 0.100.30. Conclusion: The Reliability criteria for LSE defined by the Fibroscan manufacturer are not relevant: the only parameter influencing LSE accuracy is IQR/M; number of valid shots and success rate had no influence. IQR/M defines 3 groups of LSE: “very reliable” when IQR/M is ≤0.10, especially in LSE >13 kPa; “reliable” when IQR/M is between 0.10 and 0.30; and “unreliable” when IQR/M is >0.30 in LSE ≥7.6kPa.
This site complies with the HONcode standard for trustworthy health information.
The Content on this Site is presented in a summary fashion, and is intended to be used for educational and entertainment purposes only. It is not intended to be and should not be interpreted as medical advice or a diagnosis of any health or fitness problem, condition or disease; or a recommendation for a specific test, doctor, care provider, procedure, treatment plan, product, or course of action. Med Help International, Inc. is not a medical or healthcare provider and your use of this Site does not create a doctor / patient relationship. We disclaim all responsibility for the professional qualifications and licensing of, and services provided by, any physician or other health providers posting on or otherwise referred to on this Site and/or any Third Party Site. Never disregard the medical advice of your physician or health professional, or delay in seeking such advice, because of something you read on this Site. We offer this Site AS IS and without any warranties. By using this Site you agree to the following Terms and Conditions. If you think you may have a medical emergency, call your physician or 911 immediately.