I have 2 BP monitors and they almost always are very inconsistent with reading. I'll check at the same time too.
One is the OMRON HEM-711DLX arm cuff monitor. The other is a Timex wrist monitor.
The arm cuff monitor (Omron) will say something like 121/75, while at the very same time, the wrist monitor, on my other arm will say 132/95. If my actual BP is 132/95, I'd like to know, cause that means medication.
I am sitting down when I measure my BP and I have my arm, and wrist at heart level as per instructions.
You would think the Omron arm cuff monitor would be more accurate but for some reason I suspect the wrist monitor is close to my true value. I say this because every once in a while my arm cuff monitor will say something like 99/67. And I know my BP is never that low.
P.S. I get 2 TOTALLY different reading on the wrist monitor when I lie down flat on my back and have the monitor over my heart vs. sitting up and holding it up at heart (which is what the directions call for). I've read that lying down give you a MORE accurate reading though. Very confused.
In general, arm cuff monitors are more accurate than wrist monitors.
You need to remember this:
Blood pressure is affected by gravity. If you are standing and measuring your blood pressure at ankle level, you are going to see some amazing numbers (just try), probably 220/180 or so. The explaination is fairly simple:
1 mmHg (millimeter of mercury) is equivalent to approx. 0,75 cm of water (as the density of mercury is approx. 13 times as high as water). Blood has about the same density as water, so in other words, for each cm you keep your arm (wrist) below heart level, your blood pressure monitoring will increase with 0,75 mmHg systolic AND diastolic.
If your arm is hanging straight down when measuring, you will get a wrong monitoring with your arm cuff but a severely wrong monitoring with your wrist monitor.
If you are going to measure your blood pressure, here is how to do it:
Supine: Support your arm on a small pillow so it is leveled with your heart.
Sitting: Support your arm on a table or similar.
Make sure you do not bend your elbow so much, as this may also produce wrong numbers.
Copyright 1994-2016 MedHelp International. All rights reserved.
MedHelp is a division of Aptus Health.
This site complies with the HONcode standard for trustworthy health information.
The Content on this Site is presented in a summary fashion, and is intended to be used for educational and entertainment purposes only. It is not intended to be and should not be interpreted as medical advice or a diagnosis of any health or fitness problem, condition or disease; or a recommendation for a specific test, doctor, care provider, procedure, treatment plan, product, or course of action. Med Help International, Inc. is not a medical or healthcare provider and your use of this Site does not create a doctor / patient relationship. We disclaim all responsibility for the professional qualifications and licensing of, and services provided by, any physician or other health providers posting on or otherwise referred to on this Site and/or any Third Party Site. Never disregard the medical advice of your physician or health professional, or delay in seeking such advice, because of something you read on this Site. We offer this Site AS IS and without any warranties. By using this Site you agree to the following Terms and Conditions. If you think you may have a medical emergency, call your physician or 911 immediately.