Can someone help me look online for info on whether or not it is considered "safe" to have an MRI on a 3T if you have a green filled filter/ (it's located in my vena cava)? My filter is made of titanium and can be temporary or permanent. I want to be prepared next time if and when I decide to pursue this thing. I did manage to look some things up and found that it is considered safe on a 3T but nothing bigger. I just don't know how reliable these sources are. If it's true that I could then... I got lied to and I hate that.
Most currently used clinical MR scanners are 1.5 to 3 tesla (T), which corresponds to 30 000 to 60 000 times the strength of the Earth’s magnetic field. The greatest risk from the main magnetic field is attraction of a ferromagnetic object into the scanner.
For the purposes of this statement, the term "ferromagnetic" is used to denote a substance that experiences an attractive force in the presence of a magnetic field. As a result of ferromagnetic interactions, a device may be moved, rotated, dislodged, or accelerated toward the magnet.
Thus, a ferromagnetic object might be accelerated toward the magnet at dangerously high velocities and/or with dangerously high forces, creating a "projectile effect" that could lead to significant patient injury or damage to the MR system.
Device function may also be altered or negated as a result of interactions with the strong static magnetic fields. Most, but importantly not all, currently implanted cardiovascular devices are either nonferromagnetic or weakly ferromagnetic. The higher the static magnetic field of the MR system, the greater the resultant ferromagnetic forces on weakly or overtly ferromagnetic materials.
Thus, findings from ex vivo studies at fields of 1.5 T or lower may not necessarily apply to imaging of devices at higher field strengths (eg, 3 T and higher), which produce significantly greater forces. For devices in which the ferromagnetism of the device is a significant safety concern, consideration should be given to performing the study at the lowest field strength available to reduce whatever ferromagnetic risk might be present.
Finally, all healthcare professionals are reminded that currently used MR scanners are typically superconducting and thus are always "on."
The entire article is at:
http://www.circ. ahajournals. org/cgi/content/full/116/24/2878 (remove spaces)
Hope this helps a little. I will see what else I might can find tomorrow.
This site complies with the HONcode standard for trustworthy health information.
The Content on this Site is presented in a summary fashion, and is intended to be used for educational and entertainment purposes only. It is not intended to be and should not be interpreted as medical advice or a diagnosis of any health or fitness problem, condition or disease; or a recommendation for a specific test, doctor, care provider, procedure, treatment plan, product, or course of action. Med Help International, Inc. is not a medical or healthcare provider and your use of this Site does not create a doctor / patient relationship. We disclaim all responsibility for the professional qualifications and licensing of, and services provided by, any physician or other health providers posting on or otherwise referred to on this Site and/or any Third Party Site. Never disregard the medical advice of your physician or health professional, or delay in seeking such advice, because of something you read on this Site. We offer this Site AS IS and without any warranties. By using this Site you agree to the following Terms and Conditions. If you think you may have a medical emergency, call your physician or 911 immediately.